Tagged: damaged lenses?
December 16, 2017 at 2:12 pm #29248lreinharParticipant
I had great fun putting my Deluxe Foldscope together, but neither lens is clear. Each lens has a black “amoeba-shaped” spot in middle of the field, and the rest of the field is “crazed”. Could the lenses have been damaged in the manufacturing process?
I am giving four other Foldscopes as Christmas presents, and I am afraid that the kids will be as disappointed as I am.
Can I get replacement lenses?
I am open to suggestions for how to view objects with the Foldschope.
January 2, 2018 at 6:11 am #29771AnonymousInactive
- This topic was modified 3 years, 7 months ago by lreinhar.
I’m sorry to read that your foldscope’s lenses are dirty. Have you tried to clean them up with the cotton swab ?
What about the other foldscopes ?January 7, 2018 at 2:13 pm #30000Chuck YoungParticipant
I am having a similar problem with lenses. I find that the focusing wedge is useless; the lens must be VERY close to the slide for focusing, closer than can be achieved with the focusing wedge in place.
I found an alternate way to check lens performance without assembling the entire foldscope;
see my note below. I checked the lenses on 2 classroom sets (40 lenses total).
My suspicion is that there is a problem with the production run of the lens mounts.
I doubt that (at this price) anyone tested the individual lenses.
I am also finding that the slots for the slides in the foldscope are not long enough. I had to lengthen the slots with an exacto knife.
*I will describe my attempts to check my two boxes of Basic Classroom FoldScopes.
I think my procedure is quite simple.
My light source is the overhead lamp in our dining room. I redirect the light by placing a hand mirror one the table so the light from overhead lamp is in my eyes.
I am using:
one of the fern rhizome prepared sample slides, and
one of the magnetic coupler washers and
one lens in its black holder.
I also have a package of blank microscope slides with cover slips.
I place a magnetic coupler washer on the table, black side up.
I place the prepared slide on the magnetic coupler washer, label and sample side up. I center the sample over the hole in the magnetic coupler washer.
I place the lens on the slide, silver side down. So, I have a sandwich of magnetic coupler, prepared slide and lens The magnets hold the sandwich together.
I hold the sandwich up to the light, and hold it close to my eye. Sometimes I have to readjust the sandwich to center the sample on the lens. I can see the cells of the sample. For some of the lenses, the focus is good. In this particular lens,
the focus is kind of poor. I can improve the focus by prying the lens away from the slide OR by adding one or two cover slips between the lens and the microscope slide.
I tested all 40 of the lenses on my classroom FoldScopes. Three or four of them gave the best images; very much in focus, good contrast, etc. Most gave reasonably good images that could be improved using the cover slips. A couple of them gave images that were out of focus and couldn’t be improved with the slide cover slips.August 24, 2018 at 10:23 am #57740RpvvshalimarbaghParticipant
WE IN OUR SCHOOL RECIEVED 30 FOLDSCOPES AND ALL OF THEM HAVE THE SAME PROBLEM. NONE OF THEM IS CLEAR. YES WE TRIED TO CLEAN THE LENSES BUT IS OF NO USESeptember 3, 2018 at 12:20 pm #59947donnacnhParticipant
I had this problem too, but then I looked closer at my foldscope and it had an extra magnetic coupler attached to the lens piece, once I took off the extra coupler the lens was clear since now it was closer to the slide and I could see everything easily.
Also, check to be sure that your slide is in properly, the slipcover should be down and closest to the lens, otherwise you are looking through the slide to the item and it appears to be unclean.
If you have not looked at the tutorial videos, I suggest that you do, I tried to just use the instruction book, but found the tutorials to be much more helpful.
Happy discovery!!September 7, 2018 at 12:03 pm #61156PrabhatkumarParticipant
My foldscope has too bubble like light spots.
How can I get to clear images???December 23, 2018 at 12:33 pm #108954LoisTrParticipant
Donnacnh, i logged in specifically because I thought my kit only had 2 couplers and 2 lenses that were square and not round. The lenses were attached to the couplers just as you described. Thank you for taking the time to post.April 19, 2019 at 3:23 pm #144370SkyFromFoldscopeParticipant
Please email email@example.com if you are still having issues and we would be happy to troubleshoot.
Skylar from the Foldscope TeamDecember 15, 2019 at 10:49 pm #187143TeernoParticipant
Why do I see my eye pupils while viewing? Anyone else have same problems? What can I do?December 18, 2019 at 12:05 am #187245AnonymousInactive
Please send us
1. A picture of your Foldscope from the front (hinge closed.)
2. A picture of your Foldscope from the back (hinge closed.)
3. A picture of your Foldscope with the hinge open (no slide inside.)
4. A picture of your Foldscope with the hinge open (slide inserted.)
So we may further help you troubleshoot,
Thank you so much!May 20, 2021 at 1:33 am #201225
Yes, I am also having the same problem, when I am putting a sample it just doesn’t appear, it just these black spots everywhere and are NOTHING like the sample I was using, which was a leaf, please tell me how to fix this.
FadelScopesMay 21, 2021 at 10:57 am #201302paolamrParticipant
Hi FadelScopes, can you please check if you indeed are using just one lens and one coupler? Sometimes users use two couplers instead of one by mistake (because they are attached to each other, it is hard to tell).
If you are still having issues, I am happy to help you troubleshoot. Please email me to firstname.lastname@example.org
Paola (Foldscope Team)May 23, 2021 at 12:13 pm #201360
Yes I am using one viewer coupler and one lens, and one coupler in the aperture or whatever its called, but I still don’t know what to do, I clean the lenses with a cotton bud, but that’s not enough either, I wish I could send pictures of my Foldscope so you could get the problem.
FadelScopesMay 23, 2021 at 12:14 pm #201361
Also i had three couplers separated, with the third one in my foldscope pouch
- You must be logged in to reply to this topic.